Home 　2-1 Input 6. Parameter(HP、LP)PV value high / low

▇  HP － High Point Compensation (No decimal point) PT100

1. Hold "SET" key 5sec to enter LEVEL 1

2. Set HP adjustment High Point Compensation

(HP=5
When the temperature is increased to 100 degrees
show an increase of 100 + 5 degrees
When the temperature is increased to 200 degrees
show an increase of 200 + 10 degrees
When the temperature is increased to 300 degrees
show an increase of 300 + 15 degrees
and so forth...)

▇  HP － High Point Compensation (a decimal point) 4-20 mA

1. Hold "SET" key 5sec to enter LEVEL 1

2. PV increases 5%
PV increases 10%

3. Input  4    mA = 0
Input  8    mA = 26.1
Input  12  mA = 52.4
Input  16  mA = 78.6
Input  20  mA = 105.0  ▇  LP － Low compensation (No decimal point) PT100

1. Hold "SET" key 5sec to enter LEVEL 1

2. Set LP adjustment Low Point Compensation

(HP=5
When the temperature is increased to 100 degrees
show an increase of 100 + 5 degrees
When the temperature is increased to 200 degrees
show an increase of 200 + 5 degrees
When the temperature is increased to 300 degrees
show an increase of 300 + 5 degrees
and so forth...)

▇  LP － Low compensation (a decimal point) 4-20mA

1. Hold "SET" key 5sec to enter LEVEL 1

2. Set LP adjust the low compensation,
let the temperature increase of 5 degrees

3. Input  4    mA = 5
Input  8    mA = 29.9
Input  12  mA = 54.9
Input  16  mA = 79.9
Input  20  mA = 104.9 